Confirm manual entries with ENTER. Exit the parameter level at any time by repeatedly pressing the. “Parameter” key. BioPhotometer – Short instructions. Get Eppendorf BioPhotometer Manual. Get all Eppendorf manuals!. Get Eppendorf BioPhotometer plus Instruction Manual. Get all Eppendorf manuals!.
|Published (Last):||4 September 2014|
|PDF File Size:||19.37 Mb|
|ePub File Size:||8.93 Mb|
|Price:||Free* [*Free Regsitration Required]|
Aggressive chemicals may damage both the device and its accessories. BioPhotometer plus — Operating manual 5 Operation 5 5.
For calculating the FOI you can select between 2 units in the method parameters: Service product portfolio may vary according to country. Page 9 BioPhotometer plus — Operating manual 2 Product description The BioPhotometer plus converts the measured absorbance values into concentration results.
What’s missing? Tell us about it.
BioPhotometer plus — Operating manual 8 Troubleshooting biophotomster. Sample dilution and biopyotometer light path of the cuvette See Section Page 49 Page 50 Page 51 In touch with life Your local distributor: In addition to the results the device also displays the absorbance values and some other important details, e. To ensure minimum contamination of the sample, it is essential to verify its purity by photometric measurements. Connect the printer cable to the serial printer port 4 of the photometer and tighten the locking screws.
Tell us about it.
Eppendorf BioPhotometer User Manual | 53 pages
Slide the cuvette shaft cover forward to protect the shaft against contamination. Tell us what’s missing.
Purity and homogeneity of the sample are important considerations for subsequent applications. The Eppendorf BioPhotometer D30 is the 3rd generation of Eppendorf spectrophotometers to become an established standard in the life sciences.
Cell Protein DNA Core Facilities
Page 17 BioPhotometer plus — Operating manual 5 Operation 5. Similar dimensions as classic cuvettes. Nucleic acid purification is a major application in molecular laboratories. BioPhotometer plus — Operating manual 1 User instructions 1 mwnual.
Eppendorf semi-micro Vis Cuvetteplastic cuvette for measurements in the Vis range, max.
Calculating the molar concentration Application: Up to 3 mL filling volume. In the method procedure or parameter mode this key has a different function entering the sample number or entering a decimal point biophotoketer inputting values. Recalibrate or use the calibration stored.
Glass or plastic Cuvette shaft: Thus, the two display formats complement each other perfectly to allow for an evaluation of the quality of a DNA sample. Results In addition, all important raw data pertaining to the DNA sample are displayed in table format. Electric shock as a result of penetration of liquid. Easy closing for easy handling.
For the calculation of sample measurements, the most-recently stored calibration is used. UVettes in routine packs are ideal for routine applications.
Damage from improper cleaning of the cuvette shaft. The height of the light beam in the cuvette is 1. Linear and non-linear regression Evaluation procedure of the multi-point calibration.