This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3.
|Published (Last):||1 October 2013|
|PDF File Size:||9.13 Mb|
|ePub File Size:||18.65 Mb|
|Price:||Free* [*Free Regsitration Required]|
WARNING If this product is to be energized via an external auto transformer for voltage reduction, make sure that its common terminal is connected to a neutral earthed pole of the power supply.
Making a Basic Measurement This chapter is divided into the manhal sections: To be implemented in future firmware upgrade. Using the Front Panel Entering Data When setting measurement parameters, there are several ways to enter or modify the value e4440z the active function: Refer to Figure Index personality options not in instrument power amplifiers preamplifiers preset factory types user, creating printer setup probes AC and DC product information rear panel symbols rear panel features reference level, setting reference, turning on 10 MHz rename files RF limiters Firmware Revision To view the firmware revision of your analyzer, press call Agilent Technologies regarding your analyzer, it is helpful to have this revision and the analyzer serial number available.
Used Agilent EA
Press System Reference press the softkey to toggle the reference on. Place a marker labeled 1 on the 10 MHz peak, as shown in Press Peak Search Note that the frequency and amplitude of the marker appear both in the active function block, and in the upper-right corner of the screen. Is the analyzer turned on? Installation and Setup Running Internal Alignments Running Internal Alignments Each time the analyzer is powered on, the internal alignment routine runs automatically.
EaEa manuwl, EaEa.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
In Case of Difficulty Before calling Agilent Technologies, however or returning an analyzer for serviceperform the quick checks listed in the problem. This symbol is used to indicate the OFF position of the power button. To see that trace 2 is in view mode, blank traces 1 and 3 e444a pressing.
Page 25 Item Description Data invalid indicator Status message area Key menu title Softkey menu Frequency span or stop frequency Sweep time Video bandwidth Frequency offset Display status line Resolution bandwidth Center frequency or start frequency Internal preamp Auto alignment routine status Trigger or Trace Average Display line annotation A total of 3 years of warranty service support.
Viewing Traces 1, 2, and 3 Step 2. To access the menu, press Rename 2. Use the keys to display the softkey that lists the character you want. Agilent Technologies shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance, or use of this material. Options and Accessories Ordering Options and Accessories Ordering Options and Accessories Options and accessories help you configure the analyzer for your specific applications, and enable you to use the analyzer.
To access the Copy menu, press boxes as described by the From: Front and Rear Panel Features Key Overview In other key menus, one key label will always be highlighted to show which key has been selected but the menu is immediately exited when a selection is made.
Note that the file extension is always set by the system and for this file type the extension is TRC. Creating A Directory b. Knob Increments or decrements the current value. Options And Accessories Options and Accessories This chapter lists options and accessories available for your analyzer. This symbol indicates the input power required is AC. The following table describes the four types of spectrum analyzer messages.
Page 30 Front and Rear Panel Features Key Overview In other key menus, one key label will always be highlighted to show which key has been selected but the menu is immediately exited when a selection is made. e4440q
A pound uncoupled function. Types of Spectrum Analyzer Messages The analyzer can generate various messages that appear on the display during operation.
4e440a Frequency Accuracy Additional Information Notice The information contained in this document is subject to change without notice. Agilent handheld spectrum analyzer user’s guide pages. In this example you will save a trace to the internal C: